Defender™ Bench Scale, D31PBX AM | OHAUS
Defender Indicators. EN EasyConnectTM Indicator. To connect an OHAUS EasyConnectTM scale base to an i-DT33P or an i-DT33XW EasyConnectTM. This manual contains installation, operation and maintenance instructions for the TD32PE indicator BATTERY IS TO BE REPLACED ONLY BY AN AUTHORIZED OHAUS.
Please check here to exluded your comments from future marketing material Check here to exclude your comments from future marketing materials. Please rate our performance in each area Which service are you reviewing? See Section 1. Please note приведенная ссылка times are not guaranteed. Page Chapter 4 Testing These tests are performed on the Indicator manusl a known good scale base and test mass.
Reduce load on scale. Weight reading below Underload limit. Calibration data not present. Calibration Error. Calibration value outside allowable limits. Repeat calibration using correct calibration weights. The average piece weight of the parts is small warning. Use parts with average piece weight greater than or equal to 1 division. The average piece weight of the parts is too small.
Product Images. Bench and Compact scales. The Defender Series bench scales are ideal for general weighing, simple counting, and checkweighing applications in production, packaging, warehouse, inventory, shipping, and receiving areas. The Defender Series features a simple, yet rugged tubular-frame base design and indicators with tactile keys, multicolor backlit LCD display, AC or battery operation, and flexible mounting capabilities.
Magic Spreadsheet Product Images. Features Models Documents Details. Results are clearly shown using the large, bright backlit LCD display. Page Menu Setup Back Moves backwards through the upper and middle level menus.
Backs out of a list of selectable items to the previous middle level menu. Table Press No to exit the GEO menu. End appears. Press Yes to return to the CAL menu. Press No continuously until E. Press Yes to return to weighing mode. Page Span Calibration Yes. The display shows –C– while the full calibration point is established. After successful calibration, the scale exits to the active weighing mode and displays the actual weight value.
After you press Yes, INtEr appears again. Page Interface Commands Units, up to five characters. Page Legal For Trade Lft The weights and measures official may apply a wire or paper security seal as shown below. T31P Wire Seal. T31P Paper Seal. If the Indicator shows no sign of physical damage, continue with procedures. See chart below for visual checking procedures. Visual Checking Procedures.
Page Diagnostic Guide If more than one symptom is observed, approach one area at a time, and remember that the symptoms may be interrelated. If a problem arises that is not covered in this manual, contact Ohaus Corporation for further information. Page 23 Error 8.
Overload limit. Error 8. Recalibrate scale. Error 9. Cycle power.